ASSISTENZA SERVIZIO POST VENDITA
N° Verde 800910575
Risposta entro 24/48 ore
Digital Air 3D
Behind-the-ear hearing aid (personal sound amplifier).
Indications for use
Digital Air 3D is a behind-the-ear personal sound amplifier that guarantees clear sounds and hearing stability.
Description of the product
Digital Air 3D is a behind-the-ear hearing aid that guarantees the perception of clearer sounds and its open-fit design reduces the feeling of obstruction, ensuring a higher hearing stability.
To ensure maximum hearing comfort in any situation, the Digital Air 3D has two different amplification programs, one for quiet environments and another for numerous environments.
Digital Air 3D
Specifiche Tecniche DIGITAL AIR-3D
|Parametri||Prog. 1||Prog. 2|
|MAX OSPL 90||124.2 dB||120.5 dB|
|HFA OSPL90||116.3 dB||110.7 dB|
|Peak Gain||33.4 dB||29.5 dB|
|HFA Full on Gain||26.2 dB||20.1 dB|
|Total Harmonic Distortion:|
|Eq. Input Noise Level||33.3 dB||33.6 dB|
|Frequency Range||≤200 Hz ≥ 8000 Hz||≤200 Hz ≥ 8000 Hz|
|Attack Time||0 ms||0 ms|
|Release Time||19 ms||48 ms|
|Battery Current||1.20 mA||1.08 mA|
The hearing aid (personal sound amplifier) should only be used in accordance with the instructions provided in the package leaflet.
The hearing aid (personal sound amplifier) does not restore normal hearing and does not prevent or improve damage to the auditory apparatus due to natural conditions. Allergic reactions are rare. However, consult a doctor if you experience any sensations of itching, redness, soreness, inflammation or burning in or around the ear.
In the unlikely event that some parts remain in the ear canal after the hearing aid (personal sound amplifier) is removed, immediately consult a doctor.
Remove the hearing aid (personal sound amplifier) if you are undergoing CT scans, magnetic resonance imaging or other electromagnetic testing.
Do not use alcohol to clean the hearing aid (personal sound amplifier) and/or the personalised earmolds.
The personalised tubes and earmolds should never be rinsed or immersed in water, since water droplets may remain in the tube, blocking sounds or damaging the electrical components of the hearing aid (personal sound amplifier).
USED BATTERIES MUST BE DISCARDED IN SPECIAL RECYCLING CONTAINERS
This product is an EC 0123 medical device. Read the warnings and instructions for use carefully.
Authorisation date XX/XX/XX